The result of the test is indicated by the red LED (incorrect phase sequence) or the green LED (correct phase sequence) turning on, by simply moving the sensor first on the L1 and then on the L2 phase.
|Measuring range||100÷1000VAC 50÷60Hz|
|Detection of AC voltage without contact, also on insulating sheaths|
|Phase sequence without contact also on insulating sheaths with red and green LED indications|
|Phase concordance with red and green LED indications|